physician-recommended, easy to use daily
How To Use D'OXYVA?
First, press the lever on top of your D’OXYVA delivery device next to the on/off button, then slide the door up to access the device’s liquid capsule. Fill it with clean, drinkable water. The water level should be just by the orange marker. Filling your D’OXYVA with too much or too little water might cause minor water seepage, but does NOT affect proper operation. You can use the same water for multiple applications, until level drops below marked fill line.
When closing your liquid capsule, make sure it is closed tightly; you can check by turning it upside down and shaking it. If no water comes out, you are ready to move on to the next step. Now, place your liquid capsule back and make sure the ridges are aligned with the device’s ridges.
Once in place, slide the door back down until it clicks to make sure it is secured properly. Next, slide in ONLY a D’OXYVA CO2 cartridge. Do NOT try using any other cartridge as that may cause serious damage to your device and injury to yourself! Make sure that your device is turned OFF, then unscrew your cartridge cup. Check that there is a rubber O-ring in the screw neck; it may be black or gray.
Make sure you are using a brand new D’OXYVA CO2 cartridge; it should not be punctured yet. Put your D’OXYVA cartridge in the device cartridge cup and slowly screw it tightly enough to your device to put it in place, then give it a good rapid twist. You may hear some hissing from the back hole; this is fine because your D’OXYVA cartridge is pressurized.
Press the ON button, and you will see bubbling and hear more hissing coming from your liquid capsule, meaning you are ready to start your D’OXYVA application. Make sure you hold the device in the correct angle, using the stand as a guide. Not using the correct angle may affect the delivery time and drip some water, which does NOT affect proper operation.
Insert your thumb into your device’s applicator opening, and press the applicator against your skin as needed, as long as the skin is healthy. If you are using your D’OXYVA on an open wound, make sure that the device does not touch the wound. You can still spray the vapor onto the wound.
One D’OXYVA CO2 cartridge is 1 full application and will take 4-6 minutes. You can enjoy it while sitting on your couch, watching television, at your dining table, or while in bed. You will know that it is done when the bubbling subsides.
Once you are done, turn it OFF by pushing the button fully upwards. Allow the device to cool down for at least 2 minutes before removing the used D’OXYVA CO2 cartridge, which is cold due to proper operation. Please recycle ♻ used D’OXYVA CO2 cartridges by disposing them in any metal recycling bin. Then, you are ready for your next application.
How To Start?
Follow the Standard Application Guide — Or, Get Your Personalized Plan
Reduce unwanted doctor visits, recommended and ranked top by experts!